The main task of technical committees is to prepare International Standards. Draft International Standards adopted by the technical committees are circulated to. à aiguille interne. STANDARD. ISO. Second edition. Reference number. ISO (E). Provläsningsexemplar / Preview. STANDARD. ISO. Second edition. Reference number. ISO (E). This is a free 4 page sample. Access the full version online.
|Genre:||Health and Food|
|Published (Last):||21 January 2017|
|PDF File Size:||5.95 Mb|
|ePub File Size:||18.54 Mb|
|Price:||Free* [*Free Regsitration Required]|
Copyright BIS has the copyright of all its publications.
EN ISO – European Standards
The assembly comprising at least a needle tube attached to and communicating with a needle hub. This Indian Standard has been developed from Doc No.: Pages 1 and 2 4. The number of significant places retained in the rounded off value should be the same as that of the specified value in this standard. NOTE To improve the visibility of the solution, a colorant such as red or blue food dye may be incorporated.
International Standard ISO Standards are also reviewed periodically; a standard alongwith amendments is reaffirmed when such review indicates that no changes are needed; if the review indicates that changes are needed, it is taken up for revision.
Rigid tube with one end sharpened to facilitate entry into body tissue. Blood flow into the needle hub. Round the calculated average water fiowrate to the nearest whole number of millilitres.
I Principle Water is allowed to flow through the catheter and the amount of flow is measured either volumetrically or gravimetrically. Peripheral intravascular catheter designed for the introduction of withdrawal of liquids or devices into or from the peripheral vascular system.
The fluid pathway in the needle shall be free of unintended obstructions that would prevent flashback. The catheter tube may have a single lumen or multiple lumens.
Review of Indian Standards Amendments are issued to standards as the need arises on the basis of comments. The fluid is allowed to flow into the needle, and the time taken for fluid to leak through the vent fitting is measured. If a steel tube is used, it shall connply with ISO Annex D shows examples of typical needle point geom- etries. Until that time, a manufacturer may label his product “radio- opaque” provided he can support this claim by demonstrat- ing that he has an appropriate method for showing radio- opacity.
CSN EN ISO 10555-5
At the time of publication, the editions indicated were valid. Attention is particularly drawn to the following: Page 3, subclause 4. Available from Munsell Color, N. Measure the time taken for fluid to form the first falling drop at the back of the vent fitting. Figure 1 — Typical over-needle peripheral intravascular catheter NOTE 2 At the time of publication of this part of ISOthere is no acceptable, validated test method to determine radio-detectability.
An approved test method for producing a value of radio-detectability will be established. An example of a suitable apparatus is shown in fig- ure B. Certain conventions are, however, not identical to those used in Indian 105555-5.
The details of conditions under which the licence for the use of the Standard Mark may be granted to manufacturers or producers may be obtained from uso Bureau of Indian Standards. Fitting attached to the needle tube, providing communication with its bore. Limited collection of colours.
NOTE 3 The point should be designed to be noncoring. I Principle A force is applied successively tensile and connpress- ive to the needle tube and needle hub and the tube- hub union is then examined for loosening. Other parts of this standard are as follows: Catheter designed for the introduction or withdrawal of liquids or devices into or from the peripheral vascular system.
The corresponding Indian Standards which are to be substituted in their places are listed below along with their degree of equivalence for the io indicated: Assembly comprising the cath- eter tube, catheter hub and any integral fittings.
Assembly comprising at least a needle tube attached to, and communicating with, a needle hub. Search the history of over billion web pages on the Internet.
Intravascular catheters ISO , -3, -4, -5
In this adopted standard, reference appears to certain International Standards for which Indian Standards also exist. Bureau of Indian Standards BIS is a statutory institution established under the Bureau of Indian Standards Act, to promote harmonious development of the activities of standardization, marking and quality certification of goods and 10555-5 to connected matters in the country.
Suggested colour references for opaque naaterials are given in annex C. This does not preclude the free use, in the course of implementing the standard, of necessary details, such as 10555- and sizes, type or grade designations. Fixed or removable fitting per- mitting venting of air while restricting or preferably preventing the escape of blood. Part 1 General requirements do The technical committee has reviewed the provisions of the following International Standard referred in this adopted standard and has decided that it is acceptable for use in conjunction with this standard: Add the following reference: E-5 Test report The test report shall contain at least the following information: No part of the these publications may be reproduced in any form without the prior permission in writing of BIS.
Details of which is given in National Annex A. Designation and catalogue of CCR colours. When the needle is fully inserted into the catheter unit, the catheter tube shall neither extend beyond the heel of the needle bevel nor be nnore than 1 nnnn fronn it see dinnension a in figure 1.
When tested 1055-5 accordance with annex E, fluid shall not leak out of the vent fitting within 15 s. All standards are subject to revision, and parties to agreements based on this part of ISO are encouraged to investigate the possibility of applying the most recent editions of the standards indicated below.
For the purpose 1555-5 deciding whether a particular requirement of this standard is complied with, the final value, observed or calculated expressing the result of a test or analysis, shall be rounded off in accordance with IS 2: